Calibrating the Melt Flow Index (MFI) Tester is an important step in ensuring that the equipment provides accurate and reliable test results. Here are some general steps to follow when calibrating an MFI tester:
1. Check the Temperature: The first step in calibrating the MFI tester is to check the temperature of the barrel and the die. Use a reliable temperature probe to measure the temperature, and ensure that it is within the recommended range.
2. Check the Piston Pressure: The next step is to check the piston pressure. Use a reliable pressure gauge to measure the pressure, and ensure that it is also within the recommended range.
3. Prepare the Calibration Material: Prepare the calibration material according to the manufacturer's instructions. The calibration material should have a known MFI value that is close to the expected MFI value for the sample being tested.
4. Test the Calibration Material: Load the calibration material into the MFI tester, and run the test according to the manufacturer's instructions. The MFI value obtained from the calibration material should be close to the known MFI value.
5. Adjust the Equipment: If the MFI value obtained from the calibration material is not within an acceptable range, adjust the equipment as needed. This may involve adjusting the temperature, pressure, or other settings.
6. Repeat the Calibration: Repeat the calibration process until the MFI value obtained from the calibration material is within an acceptable range.
7. Record and Document: Once the calibration is complete, record the calibration results and document the process. This information can be used to ensure that the equipment is properly maintained and calibrated over time.
It is important to follow the manufacturer's instructions for specific calibration procedures, as the process may vary depending on the equipment. Calibration should be performed regularly to ensure that the MFI tester provides accurate and reliable test results.
Copyright 2022：Qinsun Instruments Co., Limited